Herbaceous Sap Flow Sensor
Monitor sap movement in herbaceous plants
Miniature size heat pulse Sap Flow sensor ideal for smaller diameter stems and woody plants.
The Herbaceous Sap Flow Sensor uses the heat pulse velocity technique to assist with estimating plant water use, and is ideal for measurements in:
Plant Water flux
Estimate Plant transpiration
Head 45mm x 15mm;
Needles 10mm x 1.27mm diameter
Temperature Sensors: 10K Precision Thermistor
Material: Epoxy and Delrin head, stainless steel needles
Cable length: 2m standard (additional cable available.)
Sap Flow sensors are a simple, accurate and durable probe to be used in conjunction with Campbell Scientific Dataloggers.
The Sap Flow sensor requires only a Heater control interface and datalogger for proper use, but the following are recommended to improve accuracy and installation ease:
Drilling Guide: for precise spacing of needles in the tree, a drilling guide is recommended for more accuracy in spacing as well as increased ease of installation.
Size 55 drill bit for best fitting installation
Mini hex chuck to hold drill bit securely for drilling.
Reflective insulation for temperature regulation
The Heater control interface, drilling guide, drill bits, mini hex chuck, and reflective insulation are available for purchase from East 30 Sensors. See Accessories for more info.
The East 30 Sensors Herbaceous Sap Flow Sensor consists of three 10mm long stainless-steel needles spaced 6mm apart. One needle contains an Evanohm heater and the others contain a precision thermistor evenly placed at 5mm. The needles are inserted into holes drilled in the trunk of a tree with the heater placed between the thermistor. A current is applied to the heater for 8 seconds, at which time the temperature of the thermistors is monitored. The flowing sap carries the heat pulse to the sensors. The time taken for the pulse peak to reach the temperature sensors is monitored. This time is directly related to the sap flow velocity.
Cohen, Y., M. Fuchs, and G.C. Green 1981. Improvement of the heat pulse method for determining sap flow in trees. Plant, Cell and Environment 4:391-397
Green, S., et al. 2003. Theory and practical application of heat pulse to measure sap flow. Agronomy Journal 95: 1371-1379.
Pearsall, K.R., et al 2014. Evaluating the potential of a novel dual heat-pulse sensor to measure volumetric water use in grapevines under a range of flow conditions. Functional Plant Biology 41: 874-883.